If you need your Westronics power supply, control boards, 2000, 3000 or DDR10 series units tested or repaired then give us a call at
We give free estimates and offer a year warranty.
If possible be sure to use the following Repair Service Order Form when sending in a Westronics for repair. We offer special discounts to the military. To send in your Westronics for repair use the shipping address below.
A subsidiary of Thermo Instrument Systems Inc
THIS IS A 10CFR PART 21 NOTIFICATION
CONTACTS: Robert T. Agee
Quality Assurance Manager
Director, Service and Tech Support
November 16, 1999
US Nuclear Regulatory Commission
Document Control Desk
Washington, DC 20555
Subject: 10CFR Part 21 Notification, Westronics Model 1200BC Recorders
This letter is a notification of a defect associated with Westronics 1200BC recorders built under the Westronics 10CFR 50 Appendix B Program.
On or about November 11, 1999 Westronics determined that five Westronics Series 1200B Commercial Grade recorders caused a short to ground upon installation at Grand Gulf Station Unit 2. Further investigation revealed that the harness assembly drawing for DC power input (Westronics Part # HN 101168) could have been mis-interpreted regarding wire color coding/pin location during harness assembly, thereby resulting in a power to chassis ground short. Inspection of DC Power Input Wiring Harnesses for 1200B recorders revealed that of approximately 135 harnesses in stock, 118 were wired incorrectly.
There have been no reported field failures of Westronics series 1200B Safety Related recorders. Although the anomaly has been identified in Commercial Grade Recorders only, three Westronics 1200BC Safety Related Recorders installed at the following locations could be affected:
|Customer/Installed Location||Serial Number||Date Shipped|
|Entergy/River Bend Station||121BC7042||07/01/98|
|The Illuminating Co. Perry Station||122BC7047
Westronics is sending this notification to the above listed customers.
Westronics Inc.•22001 North Park Drive, Suite 100•Kingwood, TX 77339-3804
Phone: 281/348-1800• Fax: 2811348-1288
E-mail: email@example.com•http://www.westronics. com
Westronics Design Engineering has developed a test to be provided to all customers of Series 1200B Commercial Grade recorders, to determined if the defective DC Power Input Harness Assemblies are installed. Westronics will provide new DC Power Input Harness Assemblies, along with installation instructions, to all purchasers of Nuclear Safety Related Recorders. It is estimated that dedicated harnesses will be shipped within 5 working days of acknowledgment of this notification.
A follow-up report will be issued within sixty days.
Should you have any questions regarding the above, please contact the undersigned.
Robert T. Agee
Quality Assurance Manager